MRI is contraindicated except under the circumstances described below. Do not allow patients with a HiRes 90K or HiRes 90K Advantage cochlear implant to be in the area of an MRI scanner unless the following conditions have been met:
The bandaging protocol recommended by Advanced Bionics is followed when the patient undergoes an MRI procedure with the magnet left in place or,
The internal magnet is surgically removed and possibly replaced with the Magnet Insert Dummy before the patient undergoes an MRI procedure.
The external sound processor and headpiece are removed before entering a room where an MRI scanner is located.
MRI testing performed for the HiRes 90K and HiRes 90K Advantage cochlear implant indicated that image shadowing may extend as far as 70 cm2 (with the magnet removed) and 210 cm2 (with the magnet in place) from the implant resulting in loss of diagnostic information in the implant vicinity. The extent of the shadowing may be minimized by adjusting the signal parameters.
MRI testing of the HiRes 90K and HiRes 90K Advantage cochlear implant with the internal magnet in place is only available in markets where regulatory approval has been received. Contact your Advanced Bionics representative for more information.
For additional information regarding the use of an MRI scanner with a HiRes 90K or HiRes 90K Advantage device, please contact Advanced Bionics Technical Support and please read the Magnetic Resonance Imaging (MRI) for the HiRes 90K and HiRes 90K Advantage CI (9050050269) or contact Advanced Bionics Technical Support.